antibiotic residue test kit,
dairy allergy test kit
Aflatoxin M1 (AFM1-0.3) Rapid Test Strip Milk antibiotics rapid test kit
MTUSBIO® Aflatoxin M1 (AFM1-0.3) Rapid Test Kit
roduct Name】 atoxin M1 (AFM1-0.3) Rapid Test Strip pecification】
96T/kitorage Conditions】 Store at 2-8℃ in dry environment. Expiry date: 12 months tended Applicatione test strip is used for detecting Aflatoxin M1 family in raw mmingled cow milk above or below tolerance and/or safe levels. Principle】 e kit is a competitive colloidal gold immunoassay method. Sample added to sample hole, if there is Aflatoxin M1 (AFM1-0.3) in mples, it will combine with marked antibodies, preventing the rked antibodies from combining the Aflatoxin M1 (AFM1-0.3) tigens of NC nitrocellulose membrane. echnique Data】 Kit sensitivity: 0.3 ppb (ng/ml) (Final detection limit) t Contents strip ………………………… 12*8T/kit well tray and sample well with reagent. ruction …………………………..1 Milk Sample Preparation】 ollect milk sample from butch, or farm ake sure the milk has no clot. ilk at room temperature, don’t heat or freeze. horoughly mix with the reagent before testing. est Procedure】 Read the manual instruction before testing. Be sure the raw mingled milk and test kit is at room temperature. 2. Take one Tube from the box, open tube and take out decided number of test to be perform, Use the test kit within an hour beyond time may cause failure of the test. The unused test strip must be sealed and keep out from sunlight and prevent moisture. 3. Take the milk sample, drop 200ul (about 4-5drops) into the sample-well, and thoroughly dissolve the milk with reagents for 30seconds minutes till the milk presents pink color. 4. Insert the test strip into the sample-well, read the results after 5 minutes. 【Results Judgment】 Test Line- Aflatoxin M1 0.3 ppb Negative(-): control line(C) display red, test line(T) color is deeper than the control line or the same as the C line, the result shows that (AFM1-0.3) concentration is less than detection Limits or not containing (AFM1-0.3). Positive(+): control line(C) display red, test line(T) color is shallower than the control line or no color, the result shows that (AFM1-0.3) concentration is higher than detection Limits. Invalid: control line (C) doesn’t display, it means the operational process is incorrect or this test card has lost efficacy. 【Notice】 Don’t use the expired or damaged products The test card removed from refrigerator should place at room temperature and should be used as soon as possible to avoid failure when wet Don’t touch sample hole and white nitrocellulose (NC) membrane Don’t mix micropipette to avoid cross contamination Samples should be clear, not turbid particles and not clogged; if not, it will easily lead to blocking and inconspicuous color, which will affect the judgment of the experimental results.