|Features1:||Touch Screen||Features2:||Power Switch|
|Features3:||Printer||Size:||280mm × 180mm × 130mm|
food testing instruments,
food testing laboratory equipments
MICROTUS INTERNATIONAL BIOTECHNOLOGY CO., LTD.
Portable Food Safety TEST KIT READER
TABLE OF CONTENTS
1. Overview 2
2. Product Parameters 2
3. Instrument Installation 2
3.1 Graphical Representation
3.2 Instrument Installation to compute
4. Instructions for Use 4
4.1 Test 5
4.2 Records 5
4.3 Settings 7
4.4 About the Instrument 9
5. Cautions 10
6. Important Safety Tips 11
MICROTUS INTERNATIONAL BIOTECHNOLOGY
CO., LTD Portable Food Safety READER can help you to quickly and accurately complete analysis of qualitative and quantitative test of multiple test kits, while obtaining the detection of the original image to facilitate query and analysis.
2. Product parameters
Power supply: 220V ± 22V 50Hz ± 1Hz Size: 280mm × 180mm × 130mm
The main function:
· 7-inch large touch screen, easy to detect operation and view history
· Built-in printer, print test reports in time
· History You can view the original detection image and test results
· U disk can be upgraded through the import test- related information.
· USB transferable data for computer.
· Automatic identification of non-card and invalid test paper card
· Single detection time: ≤ 1 second.
Figure 1 Front view of the instrument
1, Sample Hole 2,Touch Screen
Figure 2 Back view of the instrument
1. Power switch
2. Power outlet
3. USB port 1
4. USB port 2
5. RJ45 interface
7. LIS interface (medical system dedicated)
8. Bar code gun outlet (matching special test card).
The instrument can be directly connected to the computer through the USB cable, and the product manufacturer can perform the calibration, software launching, quality control and other operations through the computer software.
HOW TO USE THE INSTRUMENT
Connect the instrument to the power cable, turn on the power for about 10 seconds to enter the main interface (graph). As shown in Figure 3, the device has 3 functions -“Test”, “Records” and “Setting”. The suggested device preheating time is 10-15
Figure 3 work interface
Note: For the trial version, the measurement, reagent management, and history buttons will automatically close at the end of the trial period. After the user has activated the activation file, the system will activate the instrument.
NOTE: When you purchase a new lot test kits or will add a new test item, import the batch of test kits or the corresponding organization file of the item into the instrument. The two methods are described below.
Step 1 Add new file “GT-108”in the flash disk, copy the project file in the file-“GT-108” (other directory are not permitted).
Step 2 Connect the GT-108 to computer via the USB port, Insert the flash disk into the USB port 1 as graph 3.1(Please change the flash disk once it is not distinguished) is shown in Figure 3.1.
the project file will be listed as graph 5, select the file, pressAdd”then go back to the main page of the instrument.